Directional earth-fault protection - SVERKER 900
The fault current is set to 80mA tosimulatethe high current generated at the intermittent earth fault. The zero voltage is set to 20V;theconfiguration ismade in13 states.The fault states and the first two memory states are set to 20ms. The memory states havetimesselected togetthetrip as close to set time 1,2s as possible. In generator configuration,set U4gen to aux voltage DC.Set “Advanced mode” in SVERKER configuration.